AP240E Ver1.32 was released.(July/22/2011)
Ver1.32 is compliant with Windows 7.
Caution for when RS232C communications cable in AP140 is used
AP240E is the data analysis program for CW240.
AP240E Ver1.32 can be downloaded free of charge.
With no registration of license(product number), it may operate as an evaluation version.
Defect Report Notice
Defect Reports of CW Viewer AP240E was added.
AP240E (Ver1.11 lower) has these limitations.
(1) It allows only period( . ) as decimal symbol.
(2) It allows only comma( , ) as digit grouping symbol.
(3) It allows only slash( / ) or hyphen ( - ) as date separator on short date format.
(AP240E Ver1.12 support to above limitations.)
Last Update July/22/2011
Defect information. :
|NO.||Content of defect||Repair|
|1||It becomes a registration error at the file name for which two periods or more are used like aaa.bbb.ccc.||Ver1.20|
|2||When current range is 3kA, current wave is not correctly displayed because current range 3kA is treated as 3A.||Ver1.20|
|3||In case of specific setting of measurement range and magnification,Y axis might not be correctly displayed. (x1/3, x1/2)||Ver1.20|
|4||Measurement data when there are power failure or data omission for a long time (more than 32768 data points) might cause a registration error.||Ver1.20|
|5||After changing the display cycle of the chart of demand energy, Maximum/minimum value doesn't change by the self adjustment.||Ver1.20|
|6||When maximum value of data is 0 and minimum value is minus, Harmonics-graph occurs an error.||Ver1.20|
|7||In case of specific setting of measurement range and magnification, even if in case of the maximum, the top and bottom of waves might not be displayed.||Ver1.20|
|8||When the measurement cycle is 15 minues and the display cycle is 15 minues, the graph and report is displayed at the cycle of 45 minutes. It is similar for 15 seconds.||Ver1.20|
|9||Measurement data when there are more than 32768 data points cause a registration error.||Ver1.20|
|10||It becomes a display error when the period of the voltage change detection data exceeds 24 hours.||Ver1.21|
|11||When the manual record file is registered the error message of “Failed in the data registration” is displayed.||Ver1.22|
|12||When you display the report of the harmonics instant data, that some degrees of harmonics are non-measurement, the measurement data order disagrees with the degree order(1st to 50th).||Ver1.22|
|13||When the Second of the Start Time is not "00", the final row might not be displayed in the daily report.||Ver1.31|
|14||When the Second of the Start Time is not "00", the Interval, the Total and the Average might not be displayed correctly in the weekly report/monthly report.||Ver1.31|
|15||If you click the title line in the list on the main window, the data sequence is changed.||Ver1.31|
|16||In case of specific versions of MSCOMCTL.OCX, the combo box list might not be displayed.||Ver1.32|
|17||Printed Weekly/ Monthly Report might differ from those displayer on the screen.||Ver1.32|
|18||When you select “Actual Value” or “Content Ratio” for the display item in the Harmonics Instant Graph tab, color settings button might not be appeared.||Ver1.32|
|19||When you resize the window of the display of the waveform graph, which remains the original size of the display of the other tabs.||Ver1.32|
Technical inquiries should be addressed to:
- TEL : +81-42-534-1413
- FAX : +81-42-534-1438
- E-Mail : email@example.com
- URL : http://www.yokogawa.com/ymi/
|1||How to customize locale information.|